Watch tutorials, webinars and informative videos about PreSens optical sensor systems.
What do I have to consider when autoclaving pH Sensor Spots?
All pH Sensor Spots need to be kept away from distilled / condensed water. If you use flasks avoid this by covering the top of the flask with aluminum foil or place it upside down during autoclaving should the sensor be glued to the bottom of the flask. Put the sensor on a Petri dish to keep it heightened and away from condensed water collecting at the bottom in other vessels.